TGBT15A_915_121 - Continued spectral line regression tests
Continued tests of new manager, new HPC software, with some config_tool fixes
Get access at 8:00am. Need to switch versions, etc.
Get started at 8:50am - receiver is not in correct location, but try running a few test scans
Scan 1 - L-band, abort - apparently turret was at X-band.
Scan 2/3 runRegressXTP - mode 10 only. All fibers (in IF rack) have failed to balance? But VEGAS ADCs look ok?
All IF rack attenuators are set to 0dB, but we are only getting ~ 1V RF power, not 3V. Everything looks fine through CLEO, Bob Simon will check.
VEGASDM is not showing the snap blocks - Ray needs to check this.
9:45am - Ray completes his work on VEGASDM
Run Mode 10 scans again. IF rack still fails to balance. VEGAS ADCs balance ok, sb1snap balance fails (at -14dBFS, not -20dbFS). Ray will investigate this.
This topic: CICADA
Copyright © by the contributing authors. All material on this collaboration platform is the property of the contributing authors.
Ideas, requests, problems regarding NRAO Public Wiki?