TPTCS14A_912_121 - test RRL Groups C-band script
execute C-band observing script from AGBT14B_431, and check C-band system temps
Copy script to RRL_w3_crrl_nopk
Scan 53 - configured wrong, and VEGAS is not producing VEGASDM data. Abort
Edit script to shorten integration times, etc.
Attempt to turn all banks off. Bank C terminates abnormally.
Turn all Managers back on. Restart VEGASDM - now ok.
Appears to be using Banks C,D,E - how is this? (shouldn't D be off?)
Still doing 5 minute scans - abort again.
Bank E isn't balanced. - Attenuators are already at zero
IF Manager is complaining about IF path to VEGAS J9 and J10 (Bank E), which didn't balance.
Galen had left an incorrect cabling file in from analog IF module tests, now back to A,C,E and G as expected
Scan 59/60 done with A,C,E,G, as expected.
But, IF Manager reports: 4972/84, 5544/1229, 6561/1250, 7397/1204 for sky/bandwidth. Bank A does not look correct.
Report this back to Melinda.
This topic: CICADA
Copyright © by the contributing authors. All material on this collaboration platform is the property of the contributing authors.
Ideas, requests, problems regarding NRAO Public Wiki?