jb-pointfocus (configures for Spectrometer) 1-5 ok
jb-spectrometer - scan 6-9 Ok
jb-vegas (with vegas.vfreq=) - scan 10-13. Scan 12 did not fill in the online filler
jb-spectrometer - Scans 14-17
jb-vegas Scans 18-21
jb-vegas-wide1 Scans 22-25
There are definite drop-outs in the total power display - drops to -31dB ~ 20 secs after the start of the scan in A and E, again about 50 secs later in E only (so it is not doppler tracking I don't think). Bank B, C and F (not in use) show it also.
Now Bank A display has no peak, the total power is systematically shifting up and down. Looks like a square wave, period 30 secs, amplitude 0.3dB. Bank C manager is doing it too, but with a shorter period (10 secs; so is Bank G).
jb-vegas-wide2 Scans 26-29
jb-vegas-wide3 Scans 30-33
Bank A, C, D, E, G all have dropouts again, none of them synchronized with each other
All Banks are doing the square wave thing now - 20 sec period, 0.3dB amplitude
A,B,D,C,E all have dropouts (to -31dB roughly but not exactly.)
Bank B does the square wave thing, but whole power level drops by 1 dB. Bank D goes up by 1dB.